Storage of 9 user-defined modes allows invocation at any time. Under each mode, Continuous Rotation Mode, Reciprocating Motion Mode, and Reverse Rotation are for options.
Woodpecker Endo Gold Endomotoris mainly used in Endodontic treatment. During root canal preparation procedure, it is used to mold & clean the root canal.
a) Customized program sequence number 1-9, totally 9 programs b) Operation mode c) Battery consumption d) Set speed e) Set torque f) File system g) Operation mode h) Torque setting i) Real-time torque
Protective function of automatic reverse: During operation, if the load value exceeds the preset torque value, the file rotation mode will automatically change to Reverse Mode. And the file would return to normal rotation mode when the load is below the preset torque value again.
a) Clockwise rotation: Load value is lower than preset torque value b)Counterclockwise rotation: Load value is higher than preset torque value. c)Clockwise rotation: Load value is lower than preset torque value again.
INDICATION: The device is suitable for shaping the root canals during root canal treatment & an advantage in replacing the hand files to rotary endodontics.
CONTRAINDICATION: Patients with implanted pacemakers (or other electrical equipment).
This equipment should be stored in a room where the relative humidity is 10% ~ 93%, atmospheric pressure is 70kPa to106kPa, and the temperature is -20°C ~ +55°C.
Avoid the storage in a too hot condition. High temperature will shorten the life of electronic components, damage battery, reshape or melt some plastic.
Avoid the storage in a too cold condition. Otherwise, when the temperature of the equipment increases to a normal level, there will be dew that will possibly damage PCB board.
Oiling of contra-angle: Only the original oil injection nozzle can be used for oiling of contra-angle. After disinfection of contra-angle and before sterilization, oiling should be conducted under high pressure and high temperature.
Screw the injecting nozzle into jet of oil bottle. (Around 10 circles)
Plug the nozzle into the end part of contra-angle, and then grease the contra-angle for 2-3s till the oil flow out of contra-angle head part.
Vertically place the end part of contra-angle or tilt the contra-angle to let go the redundant oil under gravity.
Manufacturer/Importer warranty of 1 Year against manufacturing defect
Service and Warranty are handled by the manufacturer/importer directly for this product
Flipdent will not be liable to replace the Products and the liability to replace any such Product is that of the Manufacturer/Importer
Flipdent and the Buyer acknowledge that the Products purchased/used are a subject matter of original Manufacturer/Importer’s warranties only
Any or all claims in relation to returns and replacement shall be made to the concerned Manufacturer/Importer only and Flipdent may assist the customer in registering a request to get the Product replaced or serviced.
The Buyer acknowledges that the Products once sold will not be exchanged or taken back unless the seal of the package or the Product is intact or not altered in any manner or is not tampered as the case may be.
Q1.When we need to charge the battery of Endomotor?
If the battery icon turns into display that as the picture shown, it indicates that the battery capacity is very low. Please charge.
When the battery capacity is the too low, the device may indicate “Low Battery!” and automatically power off.
Q2.How to make sure that my Endomotor is charging?
Connect the power adapter with the base. Confirm that it is well connected, and then place the handpiece into the base. If the indicator light on base turns yellow, it indicates that it is charging. If the indicator light on base turns green, it indicates that the battery capacity is enough, and there is no need to charge. After charging, please unplug the power adapter
Q3.How to switch-off the Endomotor after use?
In Standby Interface, the handpiece would automatically shut down after 3 minutes without any button-pressing operation. The handpiece will also automatically shut down while it is put into the charging base. In Standby Interface, long press setting key , and then long press Adjusting key, finally the device will automatically shut down 2s later.